Analysis and Classification of Liquid Samples Using Spatial Heterodyne Raman Spectroscopy.
Spatial heterodyne spectroscopy (SHS) is used for quantitative analysis and classification of liquid samples. SHS is a version of a Michelson interferometer with no moving parts and with diffraction gratings in place of mirrors. The instrument converts frequency-resolved information into a spatially resolved one and records it in the form of interferograms. The back-extraction of spectral information is done by the fast Fourier transform. A SHS instrument is constructed with the resolving power 5000 and spectral range 522-593 nm. Two original technical solutions are used as compared to previous SHS instruments: the use of a high-frequency diode-pumped solid-state laser for excitation of Raman spectra and a microscope-based collection system. Raman spectra are excited at 532 nm at the repetition rate 80 kHz. Raman shifts between 330 cm-1 and 1600 cm-1 are measured. A new application of SHS is demonstrated: for the first time, it is used for quantitative Raman analysis to determine concentrations of cyclohexane in isopropanol and glycerol in water. Two calibration strategies are employed: univariate based on the construction of a calibration plot and multivariate based on partial least squares regression. The detection limits for both cyclohexane in isopropanol and glycerol in water are at a 0.5 mass% level. In addition to the Raman-SHS chemical analysis, classification of industrial oils (biodiesel, poly(1-decene), gasoline, heavy oil IFO380, polybutenes, and lubricant) is performed using the Raman-fluorescence spectra of the oils and principal component analysis. The oils are easily discriminated showing distinct non-overlapping patterns in the principal component space.
['Gojani|Ardian B|AB|', 'Palásti|Dávid J|DJ|', 'Paul|Andrea|A|', 'Galbács|Gábor|G|', 'Gornushkin|Igor B|IB|https://orcid.org/0000-0002-9943-7848']
Raman spectroscopy Spatial heterodyne spectroscopy diode-pumped solid-state chemical analysis DPSS