Reliability of an injection-moulded two-piece zirconia implant with PEKK abutment after long-term thermo-mechanical loading.
Zirconia implants are appreciated in some clinical indications in light of their aesthetic appearance and good biocompatibility. The aim of this work was to evaluate the performance of a newly developed two-piece zirconia/polyether ketone ketone (PEKK) implant-abutment combination after long-term cyclic loading in a hydrothermal environment, using a new protocol adapted from two available ISO standards. Sixteen implants (n = 8/group) were embedded according to ISO 14801 and divided into two groups: implants in the Observational Group (OG) were cyclically loaded for 60 days (98 N, 10 million loading cycles, 2 Hz) in 85 °C water in a chewing simulator, while non-loaded/non-aged implants (as-received) constituted the Control Group (CG). After 4.7 million loading cycles, one OG implant fractured in the chewing simulator. The surviving implants were compared to CG implants by X-ray diffraction (XRD) to investigate potential ageing as suggested by ISO 13356, but also μ-Raman spectroscopy, Focused-Ion-Beam - Scanning-Electron-Microscopy (FIB-SEM), and load-to-fracture. Ageing was shown to have limited influence on the evaluated zirconia implant, with increased monoclinic content after loading/ageing being to a shallow transformed zone of ~2 μm at the implant surface. However, OG implants showed a significantly decreased fracture load of 751 ± 231 N (CG: 995 ± 161 N; p = .046). These values enable clinical application, but the fact that one failure was recorded during cyclic fatigue along with the significant decrease in strength after cyclic loading/ageing suggest that there may be room for further optimization of especially the PEKK abutment. Furthermore, good agreement was observed between the fracture modes of the implant that failed during the cyclic fatigue experiment and the in vivo failure of one implant during pre-clinical trials, validating the interest of the in vitro protocol used in this work to check the reliability of zirconia implant.
['Zhang|Fei|F|', 'Meyer Zur Heide|Christoph|C|', 'Chevalier|Jérome|J|', 'Vleugels|Jozef|J|', 'Van Meerbeek|Bart|B|', 'Wesemann|Christian|C|', 'Camargo Dos Santos|Bernardo|B|', 'Sergo|Valter|V|', 'Kohal|Ralf-Joachim|RJ|', 'Adolfsson|Erik|E|', 'Herklotz|Insa|I|', 'Spies|Benedikt C|BC|']
D015921:Dental Implants D019232:Dental Restoration Failure D003799:Dental Stress Analysis D007659:Ketones D008422:Materials Testing D015203:Reproducibility of Results D014025:Titanium D015040:Zirconium
Zirconia PEKK Ageing Fatigue Two-piece implants