SU-E-T-387: Evaluation of Effective Treatment Depth in XXXX Treatments with Xoft Electronic Brachytherapy.

@entity1 @entity1489 @entity92
PURPOSE: To evaluate changes in the @entity92 ( @entity92 ) and effective depth of treatment based on exerted force by applicator on the skin during treatments of @entity1489 with Xoft Electronic Brachytherapy. METHODS: To simulate compressible tissue, 5mm tissue-equivalent bolus was used. An ion chamber (Soft X-ray Chamber, PTW) and electrometer (Max 4000, Standard Imaging) were used for output measurements. Measurements were done for all available Xoft surface applicators (10, 20, 35, and 50mm cones) with plastic endcap. Fig1 shows the experimental setup. The @entity92 was measured first with no or minimal pressure of the applicator on the bolus, followed by increasing uniform pressure on the applicator applied with custom cerrobend weights. The measurements were used to calculate the effective @entity92 and effective depth. RESULTS: Force applied with the applicator was plotted against the change in @entity92 relative to the @entity92 when no force is applied. For the 10mm cone, moderate force of 5N can change the @entity92 by more than 20%, (Fig2). The effect is also pronounced for the 20mm cone, while it is minimal for the 35 and 50mm cones. Even when only moderate force is applied, the effective prescription depth can be changed by a several millimeters, which is on the order of the typical prescription depth (Fig3). CONCLUSION: Based on the results of this simulation, excessive pressure applied on the @entity1 's skin by the applicator cone can drastically alter the @entity92 and effective treatment depth. The effect is most pronounced for the 10mm cone, and to a lesser extent, 20mm, which is significant as these cones tend to be used most frequently in the clinic. Applicator placement therefore may Result in significant consequences such as excessive dose to target, severe skin reaction, permanent discoloration, skin indentation, and poor overall cosmesis upon completion of treatment.
